AD series homogeneous dispersing machine are powered by micro motors of compact-structured excitation type, it can be widely used in the mixture of low, moderate or high viscosity liquid in the lab or test workshop.
1. Small in size, can save the shipping cost and takes a little space.
2. The small lab homogenizer is portable.
3. Thought simple and small, the lab homogenizer can be used to mix various kinds of liquid, cream and paste material. It is also workable for high viscosity material.
4. The rotation speed is easy to be adjusted.
5. Height of the mixing rod/paddle is adjustable within a certain range.
6. Various homogenizer heads optional.
|Rated Voltage||220±10% VAC|
|Rated Frequency||50/ 60 Hz|
|Input Power||850 W|
|Output Power||500 W|
|Rated Torque||1.78 N·m|
|Speed Control||Stepless Control|
|Speed Range||5000 ~ 28000 rpm|
|The Material of the Operating Head||Stainless steel 316L #|
|Standard Configuration||30 G|
|Diameter of working head||φ30 mm|
|Speed Of The Rotor||22 m/s|
|Mixing Capacity (H2O)||100-8000 ml|
|Max. Suitable Viscosity||6000 cP|
|Min./Max. Immersing Height||60/170 mm|
|Size Of The Mixer (T×B×H)||215 × 310 × 720 mm|
|Permissible Ambient Temperature||Lower than 40ºC|
|Permissible Relative Humidity||Less than 90%|
|1||Heterogeneous host||1||5||Supporting and fixing frame||1|
Optional Working Head
( The operating head in the main picture is just for reference. The outlook of the operating head is different depending on your choice. )
1, Put the Base (6) on the platform, connect two supporting stand (7) by screwing them together before fixing it into the Mounting hole in the base. Set in the Supporting and fixing frame (8) on the upper end of the supporting stand, and turn the handle to fasten them.
2, Connect the short Beam (9) to the dispersing host electric motor through the Mounting bolt hole at the back of the motor, then insert the Beam into the Supporting and fixing frame, and turn the handle on it to fasten the motor on the supporting stand. The host motor can be adjusted on the Supporting and fixing frame freely.
3, When installing the operating head, follow the steps in Use and Maintenance 4-5.
·To ensure the running safety of the product, fix the key points of connections firmly and inspect it regularly.
·Before testing or operating the product, please read the Use and Maintenance carefully.
·Operate the product on the Base and NO hand operation.
·The product can't be put into use in a flammable and explosive environment.
·Ensure a good grounding for the Input power supply socket. It's strictly prohibited to connect the grounding with telephone lines, metal pipes or lightening arresters.
·There must be enough space between the end of the operating head (feed inlet) and the experimental medium container, and the liquid level of the medium must be below the overflow hole of the operating head.
·Before entering the power of this machine, please carefully check if the power switch (3) on the speed controller is turned off, speed control knob (2) is in the starting position. Avoid the Power Input of emulsified host suddenly started, personal injury caused by operator and laboratory equipment, laboratory material and other damage.
Product comes with clear instructions and is easy to use and straight forward. It is very effective and able make high quality results. Also it is not very expensive. Technical support has been outstanding.
Every time I had some problem I always received professional assistance and lots of good tips. Pretty user friendly and quick to use. It is definitely worth the money spent when compared to the time I saved.
This is probably the instrument I use the most in the lab. The instrument is very easy to use because the instructions and steps are clear. Ideal for reliable and fast results. A great cost-effective homogenizer must for any lab focused on getting homogeneous results.
I would recommend that people buy this homogenizer as it makes work easier and faster. Good quality and user friendly. It’s small as well as being nice and quiet. Have not had one problem with it. Very durable"
Excellent lab mixer, appropriate speed range and plenty of torque!