AD series homogeneous dispersing machine are powered by micro motors of compact-structured excitation type, it can be widely used in the mixture of low, moderate or high viscosity liquid in the lab or test workshop.
1. Small in size, can save the shipping cost and takes a little space.
2. The small lab homogenizer is portable.
3. Thought simple and small, the lab homogenizer can be used to mix various kinds of liquid, cream and paste material. It is also workable for high viscosity material.
4. The rotation speed is easy to be adjusted.
5. Height of the mixing rod/paddle is adjustable within a certain range.
6. Various homogenizer heads optional.
|Rated Voltage||220±10% VAC|
|Rated Frequency||50/ 60 Hz|
|Input Power||850 W|
|Output Power||500 W|
|Rated Torque||1.78 N·m|
|Speed Control||Stepless Control|
|Speed Range||5000 ~ 28000 rpm|
|The Material of the Operating Head||Stainless steel 316L #|
|Standard Configuration||30 G|
|Diameter of working head||φ30 mm|
|Speed Of The Rotor||22 m/s|
|Mixing Capacity (H2O)||100-8000 ml|
|Max. Suitable Viscosity||6000 cP|
|Min./Max. Immersing Height||60/170 mm|
|Size Of The Mixer (T×B×H)||215 × 310 × 720 mm|
|Permissible Ambient Temperature||Lower than 40ºC|
|Permissible Relative Humidity||Less than 90%|
|1||Heterogeneous host||1||5||Supporting and fixing frame||1|
Optional Working Head
( The operating head in the main picture is just for reference. The outlook of the operating head is different depending on your choice. )
1, Put the Base (6) on the platform, connect two supporting stand (7) by screwing them together before fixing it into the Mounting hole in the base. Set in the Supporting and fixing frame (8) on the upper end of the supporting stand, and turn the handle to fasten them.
2, Connect the short Beam (9) to the dispersing host electric motor through the Mounting bolt hole at the back of the motor, then insert the Beam into the Supporting and fixing frame, and turn the handle on it to fasten the motor on the supporting stand. The host motor can be adjusted on the Supporting and fixing frame freely.
3, When installing the operating head, follow the steps in Use and Maintenance 4-5.
·To ensure the running safety of the product, fix the key points of connections firmly and inspect it regularly.
·Before testing or operating the product, please read the Use and Maintenance carefully.
·Operate the product on the Base and NO hand operation.
·The product can't be put into use in a flammable and explosive environment.
·Ensure a good grounding for the Input power supply socket. It's strictly prohibited to connect the grounding with telephone lines, metal pipes or lightening arresters.
·There must be enough space between the end of the operating head (feed inlet) and the experimental medium container, and the liquid level of the medium must be below the overflow hole of the operating head.
·Before entering the power of this machine, please carefully check if the power switch (3) on the speed controller is turned off, speed control knob (2) is in the starting position. Avoid the Power Input of emulsified host suddenly started, personal injury caused by operator and laboratory equipment, laboratory material and other damage.
I have been able to learn quickly how to use this product. This product is very easy to use and the working head is well designed. The technique support we received is very good. I am sure it is going to be a very popular piece of lab equipment.
Product is very easy to use, compared to similar products made by other manufacturers. The homogenizer machine has been a very beneficial addition to our lab. In addition, our lab has been very satisfied with the customer care and technical support that has been provided!
The instrument is very easy to use and allows our lab to obtain desired results for our research. It works very well with high viscosity liquid and we have never had any trouble with the machine. This is a good improvement for the homogenizing. I would highly recommend this machine to any other researchers.
The product is very effective for our use, it performed very well and the cost quite reasonable. Their service is very good and prompt to response the service call. Would definitely recommend the homogenizer and the company to people in the similar field.
This is a great product for labtory using. Portable and light, It's really easy to clean.