Lab emulsifier caters for lab application to make emulsion, or for small batch production of cosmetics, food etc. The agitator consists of rotors & stators. When it rotates at very high speed, high shear and impact forces will exert to cut, crush and shear materials for thousands of times. Different agitators for option to meet different requirements. It suits for biochemical, physicochemical, cosmetics, health care products, food etc. liquid-liquid mixing and emulsifying, liquid-solid powder dispersing and emulsifying. An ideal choice for scientific research units, product research and development, quality control and process application.
(1) Standard emulsification agitator for media required
(2) Large power output, high speed rotation to disperse emulsification agitator
(3) SS316L/PTFE agitator, highly anti-corrosive
(4) Various speed control suitable to emulsify various kinds of materials
(5) Easy assembly of agitator, quick exchange as required.
|Rated Frequency||50/ 60 Hz|
|Input Power||850 W|
|Output Power||500 W|
|Rated Torque||1.78 N·m|
|Speed Control||Stepless Control|
|Speed Range||500 ~ 15000 rpm|
|Speed Display||x10 rpm LED|
|The Material Of the Operating Head||Stainless steel 304 #|
|Standard Configuration||70 G|
|Diameter of working head||70mm|
|Speed Of The Rotor||18 m/s|
|Mixing Capacity (H2O)||500 - 40000 ml|
|Max. Suitable Viscosity||8000 mpa.s|
|Min./Max. Immersing Height||50/236 mm|
|Size Of The Mixer (T×B×H)||215 × 310 × 720 mm|
|Permissible Ambient Temperature||Lower than 40ºC|
|Permissible Relative Humidity||Less than 80%|
Structure Diagram & Packing List
AE series lab homogenizer structure diagram
1, Put the Base (6) on the platform, connect two supporting stand (7) by screwing them together before fixing it into the Mounting hole in the base. Set in the Supporting and fixing frame (8) on the upper end of the supporting stand, and turn the handle to fasten them.
2, Connect the short Beam(9) to the host electric motor through the Mounting bolt hole at the back of the motor, then insert the Beam into the Supporting and fixing frame, and turn the handle on it to fasten the motor on the supporting stand. The host motor can be adjusted on the Supporting and fixing frame freely.
·To ensure the running safety of the product, fix the key points of connections firmly and inspect it regularly.
·Before testing or operating the product, please read the Use and Maintenance carefully.
·Ensure a good grounding for the Input power supply socket.
·The product can't be put into use in a flammable and explosive environment.
·Ensure a good grounding for the Input power supply socket. It's strictly prohibited to connect the grounding with telephone lines, metal pipes or lightening arresters.
·The work of the machine against shear emulsifying head (5) before placing experiment in liquid media is up and running. Working the bottom (feed inlet) of the head should keep enough distance to the experimental media container.
·Before entering the power of this machine, please carefully check if the power switch on the speed (3) controller is turned off, speed control knob (2) is in the starting position. Avoid the Power Input of emulsified host suddenly started, personal injury caused by operator and laboratory equipment, laboratory material and other damage.
·In order to ensure security, when the experiment required a closer look at the effect, the author should wear protective cap to prevent hair The involvement of transmission of the rotating hinge (4).
Changing Stator Steps
Use random under the wrench three nuts ② with rotary, remove the three screws ⑥ from the head on the steps of the stator base ③ outside of its stator coil ⑤, remove the stator ④ that do not apply, and then to the right of the stator in, and then installed outside the stator ring and screw, three synchronous slightly tighten the nut balanced, and hand drive fixed drive shaft ①, should not have a cyclical phenomenon of elastic.
Same structure and working principle with the industrial high shear homogenizer, perfect instrument for my research and development of new formula. The manual is useful and the customer care has been excellent.
This product is essential for the work we are conducting. It has helped us tremendously and the results are excellent. The product works just perfect and very user-friendly, the price is affordable. They provides exceptional service.
As the recommended lab homogenizer, it is a general recommendation intended to address research and pilot production. We have now been using the AE500S Series for the last few weeks, worked well and no issue.
This equipment is very effective and gives high quality reproducible results. It's easy to use and performed very well. Very good for our cosmetics emulsion formula researching and testing. I highly recommend to all scientists/laboratories.
This lab high shear homogenizer is very well made. Moreover, it arrives soon after odering and is easy to operate.
We have been using it for a while with no issues. Definitely recommend it to my friends.